We also accept wire transfers, cheques and purchase orders on approved credit. Please select an existing parts list. Accepted payment methods vary by country. Please enter a message.
|Published (Last):||19 November 2013|
|PDF File Size:||19.15 Mb|
|ePub File Size:||4.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
Using the latest transformer technology, the Clamp measures currents from mA to A. It will make you familiar with the most important safety instructions for handling the Clamp. In this Warning instruction sheet, a identifies conditions and actions that Caution pose hazard s to the user. A identifies conditions and actions that may damage the Clamp or the test instruments.
This detent action should always be present in use. If the jaws open smoothly, service is indicated and the Clamp should be inspected at the nearest Fluke Service Center. Operating Instructions Page 4: Meter Readings Note Wire size and the position of the wire within the probe jaws does not affect measurement accuracy. Multimeter readings displayed in milliamps mA can be read directly as amperes of conductor current.
Readings displayed in amperes A must be multiplied times to obtain the amperes of conductor current. This warranty does not cover fuses, disposable batteries, or damage from accident, neglect, misuse, alteration, contamination, or abnormal conditions of operation or handling.
Fluke i800 AC Current Clamp Probe
FLUKE I800 PDF
Fluke i800 AC Current Clamp